VICTOR VC2003 Function Signal Generator

The instrument is a DDS digital direct composite function generator. It is cored by CPU, employs

many large scale integrated circuits, it has the following features:

(1) High precise for frequency, high stability.

(2) Purity waveform, small distortion.

(3) Non-transition process, fast frequency change-over time: can be stable instant

(4) Non-limitation on ranges: unnecessary to changes ranges, frequency covering factor can be 10 power8

(5) Sweeping characteristics: many kinds of sweeping modes are selectable and can pause randowly, LED

will display the current frequency value; can sweep meticulously on narrow band; can set initial

frequency, terminal frequency, sweeping steps and time interval.

(6) Modulation characteristics: it has amplitude modulation function, the modulation signal can be

generated by either inner signal source or outer input, the depth is adjusted and will not be

effected by carrier wave frequency and output amplitude.

(7) Convenient keyboard operation.

(8) Stability and reliability.2. Features:(1). WAVEFORM CHARACTERISTIC: sine-wave, TTL level square wave waveform amplitude resolution: 10bit sine wave harmonic distortion: -40dBc ?frequency?1M Hz, -30dBc ?frequency?1M Hz? sine wave THD? ?1% (high resistance, 1KHz) TTL level square wave rise time: ?20ns Square wave overshoot: ?5% (2). FREQUENCY CHARACTERISTIC: frequency range: 1Hz —— 3MHz resolution: less than 1MHz: 0.01Hz ?1MHz and more than 1MHz: 0.1Hz frequency tolerance: ? ±?5×10?5?10mHz+1d? frequency stability: ?±50ppm??40?——?85?? ?use the low temperature drift crystal oscillator with PLL phase-lock circuit?. Custom-made can be done if frequency stability requires to be ±20ppm or less. (3). AMPLITUDE CHARACTERISTIC: amplitude range: 10mVp-p —— 16Vp-p ?high resistance?? 5mVp-p —— 8Vp-p ?50?? display: 3 digits amplitude error: ? ±(5%+100 mVp-p)?high resistance, 1KHz, 1—16Vp-p? amplitude stability: ?±5% ?1hr, high resistance, 1KHz, 1—16Vp-p? flatness: ? ±0.5dB??100KHz?, ?±0.8dB?100KHz—1MHz?, ?±2dB?1MHz—2MHz?, ? ±3dB?2MHz—3MHz?, ?less than 1Hz, the frequency amplitude is unstable? output impedance: 50??±10%? TTL level output characteristic?low level: ?0.3V ? high level:?4.2V ?high resistance? (4). DRIFT CHARACTERISTIC: drift range: 0V —— ±6.4V ?high resistance?? 0V —— ±3.2V ?50?? resolution: 50mV ?high resistance? drift error: ? ±?5%+100mV??1hr, high resistance, 1KHz, 10Vp-p? (5). MODULATION CHARACTERISTIC: amplitude modulation: AM modulate waveform??interior amplitude modulation? sine wave: 400Hz±10%, 1000Hz±10% 2 ranges modulation depth? 1% —— 100% modulation source? inner modulation source?outer modulation source outer modulation characteristic?input impedance?600? input level?the modulation depth is 50% at 0dB input frequency response?100Hz ——100KHz?±3dB? (6). SWEEP CHARACTERISTIC? Frequency sweep mode: linearity, logarithm During frequency sweep, the frequency and amplitude value will be displayed synchronistically, and can stop at any sweeping point. It will go on sweeping or restart after pausing. sweep rate? 0.02s ——5s/step?5-digit can be set logarithm sweeping characteristic: 100 points per 10 times frequency ranges(7) .DISPLAYING? 8-digit LED displaying with F,V, AM, SWEEP, MHz, KHz, Hz, %/s indicator and every range(8).POWER SOURCE? AC 220V/110V ±10%?frequency?50Hz/60Hz±5%(9).AMBIENT TEMPERATURE AND HUMIDITY? 0—— 40? , 3.0% —— 70% RH(10). DIMENSION?270mm×215mm×100mm(11) WEIGHT? 1.9 kg

All our VICTOR serial RF Signal Generators are provided one year warrantee by us from the date you make a purchase. An replacement or repairment will be provided by us for free when a faulty is developed within the warranty.

Note: please leave contact phone number ,so we can smooth and fast delivery.